Search found 1076 matches
Good Evening! I'm trying to do a mirror of a program but i can't obtain good results. So i came here to ask if anyone can help me doing that. Sometimes the cad isn't mirrored to the right place and, in other times, the measured elements are in another differente place. The program that i try to mir...
This is Pantec WPC-2030 controller... It is plug-n-play with CMM-Manager. Just follow the instructions in the Wenzel / Pantec install guide thread. Contact your local sales AE for configuration support.
- Thu Jun 15, 2017 1:14 pm
- Forum: General Application Support
- Topic: Position error Z Axis
- Replies: 6
- Views: 5226
This will happen if the Air Pressure is too low, air bearing is tight, machine has mechanical drag, move acceleration is too high, homing machine from outside limit(s), etc. If this happens during homing, please manually move the machine closer to center of machine volume and re-boot controller / re...
- Tue May 09, 2017 10:55 am
- Forum: UI / General Usage
- Topic: File Permissions and Protection
- Replies: 0
- Views: 3556
It is possible to configure CMM-Manager software in a manner that prevents access or allows restricted access. For example it is possible to restrict access such that user can only open and run programs, thus preventing unwanted program editing. These restrictions can be enabled as Global setting in...
- Thu Jan 12, 2017 10:45 am
- Forum: Measured Features
- Topic: Unwanted point changes
- Replies: 5
- Views: 5474
This is normal for points before the Alignment to drift in location... This is just the results of the program running on your CMM in a slightly different location each time. This would be strange to see after the Alignment (Free Position does not count) Can you confirm the Alignment is determinant?...
- Fri Jan 06, 2017 3:59 pm
- Forum: Vision / Video Measurement
- Topic: How do you "Get Tip" for vision
- Replies: 1
- Views: 3437
Just open Tip manager and Select a Tip... There are two tabs in Tip Manager if you have a vision probe. When you click the Select button, it should add a line to the program.
I cannot imagine how RPS is used with 15 datum target points... This sounds more like a Best Fit Alignment. With this said, RPS does allow for over-constraint and should work well as long as the alignment is created correctly. I would work with Help Desk on this issue. You can send the Project file ...
- Thu Dec 22, 2016 11:56 am
- Forum: Reporting
- Topic: Leverage Excel Formulas within Report
- Replies: 1
- Views: 3826
It is possible to include Excel Formula into CMM Report. Simply open original copy of report in Excel, add the formula in a Note field of the report, then copy the text from Excel back into the Note field of the original CMM-Manager report operation. The Excel Formula can be entered in as Note in Re...
Okay... Here is where I set my default tolerance(s) for Live Reporting... report_1.jpg It seems the values (range) of the graph is correct, however the coloring on the graph and the annotations seems to be strange. I'll look into this a bit further. report_2.jpg Either way the results are correct......
Len... Yes, you should adjust the default value for Cloud to CAD (3D specifically) to adjust the default tolerance used during Live Reporting. Do you really have tolerance where +/- are both on positive side of nominal? If so, I will test to see how that works with graphical report / Live Reporting....