C:\ProgramData\Nikon\CMM-Manager 3.5 SP1\Driver
Open either IQSC100.ini (for Insight IM-USB) or CaliperPCI.ini (for Caliper PCI) and add the line
Code: Select all
SCAN_RESOLUTION 1.0 //hard probe scan resolution in mm
Note: The "Hard Probe" option must be selected in Probe Assembly dialog in order to enable Scanning / Hard Probe functionality. To measure with Hard Probe just select Measure Cloud from Blue measurement toolbar and choose Scan vs. Trigger option in measurement dialog.
Note: A Foot Switch or any normally closed switch must be wired inline with current Touch Probe to enable Scanning. See manufacture for details on switch.