5 posts • Page 1 of 1
After locate the rack in UCC Server, clear calibrations then calibrate the probe I get this error message when load probe assy. Any suggestions be appreciate
You do not have the required permissions to view the files attached to this post.
When you re-calibrated rack where did you measure points? You should have moved doors to open position and measured points on metal surfaces. If you measured on plastic then this explains incorrect location of calibration artifact. Otherwise it's possible to calibrate that calibration artifact unique from rack. See the Calibration Tools tab in UCC Server.
Good try CrashN8, the picture is after re-located rack not when I am re-locate it (Instruction are very clear with pictures along with texts). And you are correct too, I need to calibrate that artifact after calibrated the rack. Thanks again CrashN8